- Home /
- Infrared Traceable Thermometer Gun
Unique design permits one-handed point-and-shoot readings. Laser-sighting beam permits aiming precisely at target surface.
- Range is –58 to 1832°F and –50 to 1000°C
- Resolution of 0.1° to 200, 1° over 200
- Accuracy is ±1.5% + 2°C.
- Field of view is 50:1 (for example, at a distance of 100 inches the reading spot size is 2 inches)
- Backlighting at the touch of a key
- Data hold freezes display reading
- Low-battery indicator
- °C/°F switchable
Multi-point calibration on an individually-numbered Traceable® Certificate which assures accuracy from our ISO/IEC 17025:2017 (1750.01) calibration laboratory accredited by A2LA. It indicates traceability of measurements to the SI units through NIST or other recognized national measurement institutes (NMI) that are signatories to the CIPM Mutual Recognition Agreement.
Battery, carrying case and Traceable® certificate
Traceable provides a wide offering of Professional Infrared Thermometers with varying features and accuracies depending on your application.
We now offer an FDA-cleared infrared thermometer. (4486)
Our other infrared thermometers are not approved for medical usage and are not FDA approved. For help in selecting an infrared thermometer please contact the customer service team at Traceable.
|Infrared Range||–50 to 1000°C (–58 to 1832°F)|
|Infrared Accuracy||±1.5% + 2°C between –20 to 200°C|
|Field of View||50:1 (at a distance of 100 inches, reading spot size is 2 inches)|
|Emissivity||Adjustable from 0.10 to 1.00 in 0.01 steps|
|Temp MIN/MAX Memory||Y|
|Exportable Data (Y/N)||N|
|LCD Digit Size||3/4 in|
|A/C Plug (Y/N)||N|
|Supplied With||battery, carrying case, Traceable® Certificate|
Check items to add to the cart or select all
High end unit at a good price. Review by Paulito
Pricing (1=Overpriced, 5=Excellent) Quality (1=Poor, 5=Excellent) Value (1=Poor, 5=Excellent)
I'm cheap when it comes to purchasing for clients. I purchased this IR Thermometer from a Control Co distributor. It was a gift for a chef. It has gotten rave reviews from him and his colleagues. I realize this unit it not truly designed for such a consumer-ish task, but it is THE best for the price. (Posted on 12/18/2014)